Yes, I agree with you but there are a few important différences of running between GW rev 2.6 and rev 2.8.
In the 2.6, it was possible to tune manually the DC offset on I and Q channels (like the gain or phase). The values was memorized and a LoadConfig allowed to keep this tune. Seems not possible with the rev 2.8.
Now, the IQ DC offset is tunable into the calibration tab, also into the RFE tab ?. But the manually tuning seems not active ??? only via the calibrate function !.
For me, the "DC corrector" (RXTSP tab) is a dynamic correction. See the LMS data sheet "RX DC Correction".
The previous DC offset (calibration tab) rans well with my application.
Now, the new DC offset and DC corrector run differently...
The manually setting of the IQ gain is still active. You can tune manually the gain. But the DC offset runs differently.
These are my remarks about the changes between both revisions.
Is it possible to know what are the differences in these calibrate and offset functions between both revisions.
Please, could you give more details or information about these changes / updates ?
I will try ASAP the LMS_WriteParam (I have to add this function into the Matlab class, not present).